r36xxproblems

# Differences

This shows you the differences between two versions of the page.

 r36xxproblems [01/12/2015 15:25]pieter [Erroneous measurements] r36xxproblems [01/12/2015 15:27] (current)pieter [Erroneous measurements] Both sides previous revision Previous revision 01/12/2015 15:27 pieter [Erroneous measurements] 01/12/2015 15:25 pieter [Erroneous measurements] 01/12/2015 15:03 pieter 01/12/2015 15:02 pieter [Measurement problems] 01/12/2015 15:02 pieter [Measurement assistance] 01/12/2015 15:02 pieter [Technical Assistance Form (T.A.F.)] 01/12/2015 15:01 pieter 01/12/2015 15:00 pieter 01/12/2015 14:59 pieter created 01/12/2015 15:27 pieter [Erroneous measurements] 01/12/2015 15:25 pieter [Erroneous measurements] 01/12/2015 15:03 pieter 01/12/2015 15:02 pieter [Measurement problems] 01/12/2015 15:02 pieter [Measurement assistance] 01/12/2015 15:02 pieter [Technical Assistance Form (T.A.F.)] 01/12/2015 15:01 pieter 01/12/2015 15:00 pieter 01/12/2015 14:59 pieter created Line 20: Line 20: * Test the connected device which causes the error for its good operation and electrical influence to the process fluid. If this cannot be solved by placing an earthing rod, revise the installation schematics to avoid current loops to the electrode. ​ * Test the connected device which causes the error for its good operation and electrical influence to the process fluid. If this cannot be solved by placing an earthing rod, revise the installation schematics to avoid current loops to the electrode. ​ * If the __error remains__, measure with a voltmeter the voltage between the **Mass** terminal and the electrode reference terminal (CHAN-): ​ * If the __error remains__, measure with a voltmeter the voltage between the **Mass** terminal and the electrode reference terminal (CHAN-): ​ - * DC = ____ V + * DC = ----- V ? - * AC = ____ V + * AC = ----- V ?